Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||25 June 2014|
|PDF File Size:||20.10 Mb|
|ePub File Size:||11.52 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hewlett-Packard B Signal Generator
Set the amplitude of the RF signal source to the same level on the spectrum analyzer as set in step You will always return to this menu when you exit the selected function.
The data entry mode is established when the Attentior bus control line ATN is false. Zero the measuring receiver and wait for the zero LED to go out. The initialized value of each stored increment for the Signal Generator is listed below: Comments If the FM peak deviation increment is set to a value that would cause the Signal Generator to exceed its peak deviation range, the step-up and step-down keys become inoperative.
Indications The output amplitude changes by the stored value of amplitude incremenL Comments Keeping either key pressed continues to change the output amplitude. Change the display to units of dB pV. Receiving the Trigger Message The Signal Generator does not have the capability to respond to the trigger message. Manual Update addition and replacement pages provide updated information pertaining to selected components.
If the Signal Generator has been repaired or the heterodyne band of frequencies do not pass, the Optional Residual FM Test steps should be performed. FM Distortion at Hz and 1 kHz rates: Maintenance described herein is performed wUh power supplied to the instrument, and protective covers removed.
Function Vac Range mV f.
Set the Signal Generator frequency to a value within the range specified in the following table and verily that the measured result does not exceed the limit specified. NOTE If the performance tests are to be nanual valid, the following conditions must be met: Set the oscilloscope as follows: Please note that Hewlett-Packard’s former test and measurement, semiconductor products h; chemical analysis businesses are now part of Agilent Technologies.
Indicator Accuracy internal rates: Provides information to allow you to use the feature or function. The Data message 865b6 is sent will cause the 7 least significant HP-IB data lines to be placed in both their true and false states. Option adds the capabilty to generate Pulse Modulation using an external signal source. The results can cause degradation of device performancet early failure, or immediate destruction.
Full text of “HP: B B final”
Output amplitude resolution is 0. Locate the selection link Wl. Press the Done key when you have completed this operation. Increment entries are checked against maximum and minimum allowable increment limits.
This will set all memory location pointers to thw default locations i. Remove Fuse Holder with fingers. Exceeding this limit may cause damage to the KP SB.
HP 8656B Manuals
Connect the digital multimeter to A13J2 pin4. These two actions combine to place the Signal Generator into remote operation.
Whenever it is likely that the protection has been impaired, the Signal Generator must be made inoperative and be secured against any unintended operation. In any correspondence, refer to the Signal Generator by model number and full serial number.
Pulse Modulation is available as an option on the HP B only. Typical total FM distortion harmonic and non-harmortic is less than 1. Only one AM depth and one FM deviation can be selected. Figure provide; some examples of Data messages. Connect equipment as 865b in Figurefor the signal path shown as below. Leave the carrier frequency set to MHz.